Rehabilitation of a congenital palatal defect with a modified technique: a case report
© Bagis et al; licensee BioMed Central Ltd. 2008
Received: 03 June 2008
Accepted: 16 July 2008
Published: 16 July 2008
Constuction of the obturator prostheses and providing the patient's comfort with them are very difficult. Primary and the most important stage for these prostheses is to perform a proper impression.
A 55-year-old edentulous Turkish female patient with a congenitally maxillary defect was rehabilitated with an open hollow obturator prosthesis. After the preliminary impression was completed, a metal frame suitable with the maxillary defect was constructed manually and used for making the impression of the defect area. After the first part of the obturator was finished, second part which separates nasal cavity and oral cavity was constructed by the aid of the bulb.
When constructing an obturator prosthesis, making a detailed impression from the defect area can be performed by the aid of a metal frame, and an intraorally shaped extension that separates oral cavity from nasal cavity might be more effective for adaptation of the prosthesis.
Maxillary defects can be sourced by congenital malformations or the acquired defects resulting from surgeries . The maxillofacial-prosthodontist has two primary objectives for rehabilitation of the maxillary defects. These objectives can be explained as to restore the functions of mastication, deglutition, and speech and to achieve normal oro-facial appearance . Defect of the maxilla, which occurs as a result of tumor surgery or congenitally, may be closed with an obturator, which is a disc or plate, natural or artificial. The prosthesis that restores the maxillary defect is termed as a maxillary obturator. . The obturator prosthesis that close these defects and separate the oral and nasal cavities may be constructed in different sizes and shapes, depending on the extent of the defect . Oburators are classified as solid, open hollow and closed hollow as to the nature of their extension into defect site [5–10]. An open or closed hollow obturator is usually preferred to reduce the weight of the prosthesis when the maxillary defect is large, . Both open and closed hollow obturators allow for the fabrication of a lightweight prosthesis that can be tolerated by the patient while effectively extending into the defect. Closed hollow obturators are usually fabricated with acrylic resin (polymethyl metacrylate resin), which is porous and able to absorb water [11–13], so consequently the use of a closed hollow obturator is often avoided [6, 9]. The reduced weight, better hygiene conditions, easy fabrication and increased speech ability are some of the advantages of using an open hollow obturator . The retention of the maxillofacial prosthesis will vary with the size and configuration of the defect, the amount and contour of the remaining palatal shelf, height of the residual alveolar ridge, the size, contour, and lining mucosa of the defect and the availability of undercuts . A stable and retentive obturator fabrication has some difficulties without the assistance of teeth or implants for additional support .
This clinical report describes the prosthetic rehabilitation of an edentulous patient with congenital maxillary defect with an obturator made by a modified technique to restore the defect and separate the oral and nasal cavities from each other.
A 55-year-old edentulous Turkish female patient was referred to the Department of Prosthetic Dentistry in Karadeniz Technical University for examination and treatment. The patient had a history of congenital palatal defect with an opening between oral and nasal cavities. Detailed case history revealed that the oronasal opening was present since year of birth and the defect was not treated surgically. The patient's major complaint was being edentulous after looosing her natural teeth because of periodontal diseases. She had never had obturator prosthesis until she lost her last teeth. Construction of obturator prosthesis was decided for the rehabilitation of the patient after clinical examination.
The prosthetic rehabilitation of an edentulous patient with congenital maxillary defect with an obturator has some difficulties at the stages of impression and construction. Even the defect area is filled with a tampon before the impression, protecting the soft tissues from the residual impression materials and making a detailed impression is not easy. In this respect, recording the tissue borders with modeling wax around a metal frame intraorally may be more protective and still an effective method. Separating nasal and oral cavities from each other helps to form speech voices better and to protect nasal cavity from the food escape during chewing, and swallowing. Constructing a second part which extends to the palatal defect cavity on previously constructed bulb using an intraorally shaped modeling wax, helps the second part to cover the defect area without any damage to the soft tissues even during the functional movements. Fabricating method of this open hollow obturator provides better initial record base retention, which is also compatible with the oral tissues. And the design of this obturator prosthesis has also some advantages as: better phonation and protection of food escape to nasal cavity.
Fabricating a successful obturator prosthesis used for the prosthetic rehabilitation of congenital or acquired defects in maxilla depends on making a detailed impression and constructing the prosthetic parts compatible with the oral tissues. This clinical report describes an intraoral technique for impression making and fabrication of open hollow obturator prosthesis.
Written informed consent was obtained from the patient for publication of this case report and accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
- Keyf F: Obturator prostheses for hemimaxillectomy patients. J Oral Rehabil. 2001, 28: 821-829. 10.1046/j.1365-2842.2001.00754.x.View ArticlePubMedGoogle Scholar
- Beumer J, Curtis TA, Firtell DN: Prosthodontic and surgical considerations. Maxillofacial Rehabilitationw. 1979, The C.V. Mosby Co., St. Louis, Toronto, London, 188-243.Google Scholar
- Chalian VA, Drane JB, Standish SM: Multidisciplinary practice. Maxillofacial Prosthetics. 1971, The Williams & Wilkins Co., Baltimore, 131-148.Google Scholar
- Oral K: Construction of a buccal flange obturator. J Prosthet Dent. 1979, 41: 193-197. 10.1016/0022-3913(79)90307-X.View ArticlePubMedGoogle Scholar
- Schneider A: Method of fabricating a hollow obturator. J Prosthet Dent. 1978, 40: 351-10.1016/0022-3913(78)90046-X.View ArticlePubMedGoogle Scholar
- Phankosol P, Martin JW: Hollow obturator with removal lid. J Prosthet Dent. 1985, 54: 98-100. 10.1016/S0022-3913(85)80080-9.View ArticlePubMedGoogle Scholar
- Wu Y, Schaaf NG: Comparison of weight reduction in different designs of solid and hollow obturator prostheses. J Prosthet Dent. 1989, 62: 214-217. 10.1016/0022-3913(89)90317-X.View ArticlePubMedGoogle Scholar
- Wang RR, Hirsch RF: Refining hollow obturator base using light-activated resin. J Prosthet Dent. 1997, 78: 327-329. 10.1016/S0022-3913(97)70035-0.View ArticlePubMedGoogle Scholar
- Mc Andrew KS, Rothenberger S, Minsley GE: An innovative investment method for the fabrication of closed hollow obturator prosthesis. J Prosthet Dent. 1998, 80: 129-132. 10.1016/S0022-3913(98)70098-8.View ArticleGoogle Scholar
- Asher ES, Psillakis J, Piro JD, Wright RF: Technique for quick conversion of an obturator into a hollow bulb. J Prosthet Dent. 2001, 85: 419-420. 10.1067/mpr.2001.114276.View ArticlePubMedGoogle Scholar
- Anusavice KJ: Philips Science of Dental Materials. 1996, WB Saunders, 254-262. 10Google Scholar
- Craig RG, Powers JM: Restorative Dental Materials. 2002, St. Louis; Mosby, 636-682. 11Google Scholar
- Barrie JA: Water in polymers. Diffusion in Polymers. 1968, London:Academic Press, 259-31.Google Scholar
- Habib BH, Driscoll CF: Fabrication of a closed hollow obturator. J Prosthet Dent. 2004, 91: 383-385. 10.1016/j.prosdent.2004.01.020.View ArticlePubMedGoogle Scholar
- King GE, Martin JW: Complete dentures for the obturator patient. Dent Clin North Am. 1996, 40: 217-237.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.